Errata for: D. Prutchi and M. Norris, Design and Development of Medical Electronic Instrumentation – A Practical Perspective of the Design, Construction and Test of Medical Devices, ISBN: 0-471-67623-3, 450 pages, John Wiley & Sons, Inc., Hoboken, NJ, November 2004.
Pages 86, 87:
October 18, 2005 – Bob Higgins brought the following to my attention:
“…On page 86, the slew rare limit bridge values appear to limit SR to 333V/s. The largest gain before the bridge is 500, so the limit appears to be 0.66V/S applied to the ECG signal, but earlier in the book, 0.33V/s is the recommended limit to block pacemaker pulses…” Continue reading